<<< Back to Results
Overpressure of an Explosion-Proof Enclosure
Incident Date: 15 May 2017 Time: ---
Type of Activity: ---
Cause: Pressure release
An analyzer engineer was fatality injured when the engineer removed the cover on an explosion-proof enclosure as part of a routine task. The threaded cover, weighing ~12 lbs. and measuring 14 inches in diameter, was forcefully propelled from the enclosure as the engineer unscrewed it, inflicting a fatal head injury. The force was caused by pressure inside the enclosure from leaking sample gas or instrument air components. The enclosure was not equipped with an external indicator to determine the pressure inside the enclosure and had no means to relieve internal pressure.
Corrective Actions and Recommendations:
Figure 1: Re-enactment of cover removal
Figure 2: Example of Explosion-Proof Enclosure
Safety Alert number: 288
Whilst every effort has been made to ensure the accuracy of the information contained in this publication, neither the IOGP nor any of its members past present or future warrants its accuracy or will, regardless of its or their negligence, assume liability for any foreseeable or unforeseeable use made thereof, which liability is hereby excluded. Consequently, such use is at the recipient's own risk on the basis that any use by the recipient constitutes agreement to the terms of this disclaimer. The recipient is obliged to inform any subsequent recipient of such terms.